Have donor provide urine sample and let stand until it’s room temperature. Open the foil pouch and remove the test device & eye dropper.
- Using the enclosed dropper, withdraw the urine sample and add 4 drops into the sample oval shaped well on the cassette.
- Results are available in within 10 minutes.
List of Drugs
The list of drugs detectable with this test are as follows:
AMP- Amphetamine, 1000 ng/ml – Adderall/Ritalin
COC– Cocaine, 300 ng/ml – Crack
mAMP– Methamphetamine, 1000 ng/ml – Meth/Crystal/MDMA
OPI– Opiate, 2000 ng/ml – Heroin/Morphine/Codeine
THC– Marijuana, 50 ng/ml – Cannabinoids/Weed/Pot
The multi Drug Screen Test Device is an immunoassay based on the principle of competitive binding. Drugs which may be present in the urine sample compete against their respective drug conjugate for binding sites on their specific antibody.
During testing a urine sample migrates wiithin the cassette by capillary action. A drug, if present in the urine sample below its cut-off concentration, will not saturate the binding sites of its specific antibody. The antibody will then react with the drug-protein conjugate and a visible colored line will show up in the test line region of the specific drug cassette. The presence of drug above the cutoff concentration will saturate all the binding sites of the antibody. Therefore, the colored line will not form in the test line region.
- The test is CLIA Waived indicating the test is certified accurate and designed to eliminate mistakes during home use.
- Compared to dip panel tests, cassettes have much less opportunity for procedure mistakes because dip panel/strip tests could be under or over dipped and provide an incorrect test result. When you purchase our cassette, which includes a pipette (dropper), you’re able to deliver the exact amount of urine needed for accurate results.
Positive: One colored line appears in the control (C) region.
Negative: One colored line appears in the control (C) region, and another colored line in the test (T) region. The color intensity of the test line may be weaker or stronger than that of the control line, but regardless of the intensity, the result is negative.
Invalid: No line appears in the control zone “C”. If the test device does not produce a band in the control region, check the testing procedures, samples, and/or control materials and repeat the test using a new device.