- Perfect for testing: Employees; Patients in Abuse Centers; Students in Schools and/or Son/Daughter in the privacy of your Home
- Simple, accurate & inexpensive urine test
- FDA Cleared
- Easy to read and understand results, with 99% accuracy
|510 K #|
Have donor provide urine sample and let stand until it’s room temperature. Open the foil pouch and remove the test device & eye dropper.
The list of drugs detectable with this test are as follows:
AMP- Amphetamine, 1000 ng/ml – Adderall/Ritalin
BZO– Benzodiazepines, 300 ng/ml – Valium/Xanax/Paxam/Victan
COC– Cocaine, 300 ng/ml – Crack
mAMP– Methamphetamine, 1000 ng/ml – Meth/Crystal/MDMA
OPI– Opiate, 2000 ng/ml – Heroin/Morphine/Codeine
THC– Marijuana, 50 ng/ml – Cannabinoids/Weed/Pot
The multi Drug Screen Test Device is an immunoassay based on the principle of competitive binding. Drugs which may be present in the urine sample compete against their respective drug conjugate for binding sites on their specific antibody.
During testing a urine sample migrates wiithin the cassette by capillary action. A drug, if present in the urine sample below its cut-off concentration, will not saturate the binding sites of its specific antibody. The antibody will then react with the drug-protein conjugate and a visible colored line will show up in the test line region of the specific drug cassette. The presence of drug above the cutoff concentration will saturate all the binding sites of the antibody. Therefore, the colored line will not form in the test line region.
Positive: One colored line appears in the control (C) region.
Negative: One colored line appears in the control (C) region, and another colored line in the test (T) region. The color intensity of the test line may be weaker or stronger than that of the control line, but regardless of the intensity, the result is negative.
Invalid: No line appears in the control zone “C”. If the test device does not produce a band in the control region, check the testing procedures, samples, and/or control materials and repeat the test using a new device.
There are no reviews yet.